Wireless 1 Channel
- Effective in helping to manage particular pain problems.
- Relieving soreness in the applied area.
- Helps to strengthen and tone muscles.
- When the unit starts operation or the program has been changed. The intensity level ("density") will return to "0" .
- When the unit starts operation, the intensity ("density") output has been designed with a gradual 3 seconds raise ramp.
- When timer has reached 0 min. Unit will auto-shut down. The unit can remember last setting.
- If intensity levels ("density") are at "0" and unit is not being used, it will automatically shut down after 3 mins.
- P1~P0 timer is programmed 30 mins with backward count down.
- Screen will exchange display density/mode/timer every 3 secs.
|Size HT329L1||53 x 30 x 10 (mm)|
|HT329L2||53 dia x 10 (mm)|
|HT329L3||53 x 35 x 10 (mm)|
|HT329L4||54 x 40 x 10 (mm)|
|Channel||single channel output|
|Power supply||3V CR2025 battery 1 pc|
|Output||0~90mA P to P into 500Ω load|
|Density(Intensity)||change of 15 steps pulse amplitude|
|Timer||30 minutes to auto turn off|
|Wave form||dual phases square wave|
|Display||LCD display with timer/mode/density ("density"=intensity)|
|Mode||Part A P1~P5 are 5 TENS massage programs|
|Part B P6~P0 are 5 EMS fitness programs|
|Contraction(on time)2sec. or 4sec.|
|Relaxation (off time)2sec. or 4sec.|
|Pulse wave form|
Electrode & belt set